Service Guide. Publication Number (order as manual set) The security code is set to “HP” when the function generator is. Note: Unless otherwise indicated, this manual applies to all Serial Numbers. The Agilent .. For information on using the Phase-Lock Option for the HP A. Microsoft® Windows® included. Agilent A. 15 MHz Function /. Arbitrary Waveform Generator. Note: Unless otherwise indicated, this manual applies to all .
|Published (Last):||6 November 2018|
|PDF File Size:||19.19 Mb|
|ePub File Size:||5.12 Mb|
|Price:||Free* [*Free Regsitration Required]|
HP 33120A User Manual
The following steps show you how to enter a number in the menu. This information is mankal directly related to waveform generation but is an important part of operating the function generator.
The number of frequency points in the sweep depends on the sweep time you select and is automatically calculated by the function generator. Page 46 Chapter 3 Features and Functions Output Configuration Output Configuration This section contains information to help you configure the function generator for outputting waveforms.
See the next page if you need to change the power-line fuse. Page 6 Quick Start Page 88 Chapter 3 Features and Functions Frequency Sweep Sweep Trigger Source In the triggered sweep mode, the function generator outputs a single sweep each time a trigger is received.
The default is 1 kHz. Whether you are operating the function generator from the front panel or over the remote interface, this chapter will be useful. Chapter 2 Front-Panel Menu Operation A front-panel menu tutorial Menu Example 3 Some commands in the menu require that you enter a mannual parameter value.
HP A USER MANUAL Pdf Download.
To print the manual completely, please, download hp33120q. Bench-top viewing positions Carrying position When you enable the burst mode, the previous modulation mode is turned off.
The function generator calculates the average of the data points and compares the average to zero volts. The default is 10 Hz. The HP A is a high-performance 15 MHz synthesized function generator with built-in hpp33120a waveform capability. The following steps show you how to store and recall a state. At power-on, the units for output amplitude are volts peak-to-peak.
You can configure the function continued generator to automatically recall the power-down state when power is restored. Page Chapter 3 Features and Functions Calibration ,anual To Change the Security Code To change the security code, you must first unsecure the function generator, and then enter a new code.
The Front-Panel Menu at a Glance T he menu is organized in a top-down tree structure with three levels. Enables single trigger and triggers the generator. The default amplitude is mVpp into 50 ohms for all functions.
Chapter 2 Front-Panel Menu Operation To turn off the comma separator To turn off the comma separator The function generator can display values on the front panel with or without a comma separator.
Also notice that the hop frequency is displayed with fewer digits than the carrier frequency. Be sure to read the security code rules on page before changing the security code.
From the front panel, you can save the waveform using one of the following names: Chapter 3 Features and Functions Amplitude Modulation AM Modulating Waveform Shape The function generator will accept an internal modulating signal, an external modulating signal, or both.
HP 33120A Manuals
When the gate signal is true, the function generator outputs a continuous waveform. Chapter 3 Features and Functions Triggering Single Triggering In the single trigger mode front panel onlyyou can manually trigger the function generator by pressing the front-panel key.
Page 20 Front-Panel Menu Operation Chapter 3 Features and Functions System-Related Operations System-Related Manhal This section gives information on topics such as power-down recall, self-test, error conditions, and front-panel display control.
Duty cycle is specified as a percentage and represents the amount of time per cycle that the square wave is high. The duty cycle is auto- matically adjusted if you select a frequency that is not valid with the present duty cycle.
Enter text from picture: For certain applications where you may not want to output the sync signal, you can disable the SYNC terminal. The following steps show you how to change the amplitude to 50 mVrms. Decrements Moves majual flashing digit to the right Page 66 Chapter 3 Features and Functions Amplitude Modulation AM Modulating Source The function generator will accept an internal modulating signal, an external modulating signal, or both.